* Required fields
Please select the installation location for the belt scale matching your local conditions.
The degree of belt inclination must be such that there is no relative movement of the conveyed bulk material (material must not roll back on the belt).
actual values on siteuser input
measurement range = belt speed * 1/s =
zone of influence = 3 * measurement range =
minimum distance = belt speed / 3 * 1/s * E =
actual transition lengthuser input
Safe distancemin = 2,5 * E =
Consistent troughing and a low idler runout tolerance must be guaranteed.
The belt scale is installed in a conveyor that is protected from wind, weather and extreme temperature variations.
The belt scale installation site must be easily accessible during installation works.
Belt speed fluctuations have a detrimental effect on the belt scale accuracy.
Legal-for-trade belt scales require strict observance of the Verification Ordinance.
The final scale accuracy can only be established by in-service measurements.
The accuracies listed can only be achieved by strictly observing the regular maintenance intervals as indicated in the Operating Manual.